For DNA/RNA analysis and sample preparation
Features of TS-100 meet the increased requirements of the user, including:
- Quickly reaches the set mixing speed and maintains same amplitude of rotation around the block;
- Stable maintenance of the temperature of a wide range over the entire surface of the block;
- LCD display shows the set and actual temperature, speed and time;
- Quiet engine operation, compact size, long service life.
Heating source is a printed heating board (12 V). Mixing is provided by movement of orbital type.
The instrument is applicable in:
- Genetic analysis — in extraction of DNA, RNA and further sample preparation
- Biochemical study of enzymatic reactions and processes
- Extraction of metabolites from cellular material
Temperature Calibration Function
With the help of the temperature calibration function the user can calibrate the unit approx. ±6% of the selected temperature to compensate differences in the thermal behaviour of tubes from different manufacturers.
Temperature setting range
+25°C ... +100°C
Temperature control range
5°C above ambient ... +100°C
Temperature setting resolution
Temperature accuracy at +37°C
Average heating speed from +25°C to +100°C
Temperature uniformity over the block
Temperature calibration coefficient range
0.936...1.063 (± 0.063)
Speed control range
250 – 1400 rpm
Microprocessor controlled temperature, mixing speed and operation time
LCD, 16 x 2 signs
Digital time setting
1 min – 96 hrs (1 min increment)
Maximum continuous operation time
max. 96 hours
Overall dimensions (W×D×H)
205 x 230 x 130 mm
Input current/power consumption
12 V, 3.5 A / 42 W
External power supply
Input AC 240 V; 50/60 Hz;
Output DC 12 V
20 and 12 microtubes
0.5 and 1.5 ml
20 and 12 microtubes
0.2 aund 1.5 ml
96-well microplate (0.2 ml) for PCR
Are there any blocks for Deep well plates?
No, there are no blocks for deep well plates for this instrument. TS-100 is designed to provide high temperature uniformity, stability and accuracy in microtubes and PCR plates. Please, see TS-DW, Thermo-Shaker for deep well plates.
Is any block included in the standard supply set?
No, we offer 5 blocks you can choose from to select best suited for your application. Blocks should be ordered separately. Choose blocks at Technical Data.
How long will it take for the instrument to heat up from room temperature to maximum temperature?
It takes around 15 minutes.
Does it have a temperature calibration function?
Yes, it does. Calibration procedure is described in the user manual point 6.
Please fill in at least the fields with a red asterisk.